In the wake of the Covid-19 lockdown, there have been clear guidelines issued to businesses on keeping staff and customers safe. These include wall signs encouraging people to follow hygiene guidelines through to purpose-made floor tape to help enforce social distancing rules. For successful long-lasting application to surfaces, they all rely on the stickiness provided by the humble self-adhesive.
The route to quality
Plastic medical tubing is central to the provision of effective health care in the 21st century. It is essential for the delivery of blood, nutrients and gases to the patient, and for the execution of minimally invasive surgical procedures such as cardiac angioplasty, which have proliferated widely over recent years. This has driven manufacturers to develop increasingly more sophisticated tubing products within shorter time periods to market.
Meticulous attention to form and function in development, supported by consistent, high quality production is critical to the success of the device and the safe and effective treatment of the patient. Force testing in the development of medical tubing provides a simple, yet accurate method of qualifying the device’s important mechanical characteristics; it aids optimisation of performance and functionality and conformance to international standards.
In production, it provides a quick and inexpensive method of ensuring consistent quality, preventing faults that could have potentially lethal consequences, and damaging legal and financial repercussions for the brand. This article examines four force testing procedures for plastic medical tubing and explores the benefits and methodological considerations of each.
Since the late 1980s percutaneous coronary intervention has given cardiac surgeons an extremely effective and minimally invasive method of clearing an artery blockage, and significantly reduced trauma and recovery times over previous surgical methods.
The procedure involves threading a guide catheter through the tortuous network of arteries from the arm or groin to the treatment site. A thin, flexible metal guide wire is then threaded through the catheter and extended past the blockage.
A second, smaller catheter with a balloon attached to the tip is then threaded over the guide wire and inflated in the narrowed section of the artery to clear the blockage, and if required, place a stent in the vessel to hold its walls open. The ease with which the guide catheter may be navigated through the arteries, its trackability, incorporates a variety of parameters.
These include the lubricity of the catheter’s external surface and the forces exerted by the catheter tip at crucial junctures on route to the affected region.
Force testing in the development of these devices allows designers to minimise the forces exerted on delicate artery walls and thereby reduce the risk of perforation.
Sophisticated, custom-engineered trackability testing systems comprise a two or three dimensional model of the artery network, and an interconnected system of delicate loadcells that detect the forces exerted on the synthetic vessel walls as the guide catheter is fed through the model.
More simple and fundamental tests, however, may be performed with less sophisticated equipment.
For example, simple insertion and extractions tests of small samples of catheter into small segments of synthetic artery may be performed on a relatively inexpensive force testing system in early feasibility studies.
Similarly, insertion/extraction testing of a sample length of guide wire into the guide catheter can be used to assess the frictional forces experienced as the two materials slide over each other.
Further tensile testing of the guide wire itself will enable developers to produce as lightweight and manoeuvrable a device as possible; one that still retains sufficient strength to push through particularly severe or heavily calcified lesions within the artery. Alternatively, the coefficient of friction of the catheter’s external surface may be easily determined on an automated force testing system with a specialised fixture, aiding comparisons in lubricity of alternative potential device materials.
These techniques are not, of course, restricted to coronary angioplasty, and similar tests are suitable for catheters designed for other stent placement procedures, including peripheral and renal angioplasty.
A simple tensile test of medical tubing can provide much useful information to the developer in modelling the device’s expected performance in situ. In production, it may be employed as a quality control method for early identification of problems in the extrusion process.
There is an ever-increasing demand to decrease the wall thickness of tubing to reduce the trauma caused to delicate body tissues, provide more internal space through which fluids may flow or ancillary instruments may be introduced, and to reduce costs in mass production. Tensile tests can assess the impact of reductions in material usage on the strength and performance of the tubing.
Mapping the elastic region of the tubing’s stress-strain curve in a tensile test enables the calculation of Young’s Modulus, a useful parameter in predicting material performance. Other typical measurements include yield strength, the load at which permanent deformation of the material occurs; and perhaps most fundamentally, tensile strength, the load at which the tubing breaks altogether. Conversely, the percentage elongation of the tubing sample at a predetermined force, or at breaking point, may be of most interest.
To perform the test, each end of the tubing is wrapped around an upper and lower circular bollard grip mounted on an automated force testing system, and a tensile load is applied at a slow and steady rate until the required parameter has been observed. For routine production floor tensile strength testing, a simple console-controlled force testing system is sufficient. For more complex mapping of the material tensile properties, a computer-controlled system is normally used to enable more sophisticated collection of results, and graphical display of the tubing’s tensile profile.
It should also be noted that an extended crosshead travel may be required on the chosen system to effect sufficient elongation of larger samples.
Medical tubing joins to terminal connector components by a variety of methods, both permanent such as bonding and moulding, or reversible such as barbed luer fittings. The disassembly force of the tubing from the terminal is a critical parameter in the quality and performance of the device.
The force must be sufficiently high to avoid failure in use and ensure an hermetic seal in the joint. In the case of the reversible fitting, however, the assembly force must also remain sufficiently low to ensure ease of use in time-restricted situations.
To perform the disassembly test, the connector terminal should be held in the upper fixture of an automated force testing system; a specialised mounting jig may be required to secure the sample effectively. The tubing should then be secured to a circular bollard grip from beneath. This will spread the load evenly throughout the tubing and avoid premature failure at the gripping face and sample slippage during testing. An axial tensile load should then be applied at a constant rate until the tubing has become completely dislodged from the connector.
To assess the integrity of the seal within a luer joint, the tubing is pushed over the barbed taper with a predetermined force. Compressed air is then run through the tubing system and the joint placed under water and visually inspected for bubble formation to indicate leakage.
The packaging that tubing products are delivered in should be developed and manufactured with almost as great an attention to quality and functionality as the device itself.
Well designed and consistently produced packaging will protect the tubing and maintain sterility during transportation and storage, whilst remaining easy to open when required.
Tensile strength and elongation tests, similar to those described above, may be performed on packaging material in accordance with ISO EN 11607-1, Packaging for Terminally Sterilised Medical Devices, and EN 868-1, Packaging Materials and Systems for Medical Devices which are to be Sterilised, to guarantee conformance to minimum strength requirements. Similarly, pierce and rupture testing of airtight foil and film blister packs provide an indication of resilience to applied forces that may be encountered in the pack’s life cycle. The peel strength of adhesive seals, normally tested on an automated force testing system fitted with a peel table fixture, enables quantification of the ease with which the pack may be peeled open.
Finally, the co-efficient of friction of the packaging inner and outer surfaces may be easily determined using the same system, fitted with another specialised fixture. This aids optimisation of form-fill-seal machinery settings in processing and provides an indication of how easily the device may be slid out of its packaging on opening.
This article aims to demonstrate that one relatively inexpensive automated force testing system enables a vast range of critical quality testing procedures in the development and production of medical tubing. From tensile strength and elongation testing, to lubricity and joint integrity, investment in the correct instrumentation allows quick, easy and yet precise assessment of the functionality and quality of tubing devices, and is an essential ingredient in the delivery of successful products in today’s fast-moving technological climate.
Mecmesin's materials testers
Testing the physical strength properties of materials, in accordance with international or in-house standards, is a key part of determining their characteristics to better understand the effects of process improvements. With this in mind, Mecmesin has designed a range of static-load Materials Testers powered by intuitive VectorPro MT software.
Mecmesin OmniTest machine demo by the Duchess of Sussex at University of Chichester Tech Park opening
The Duke and Duchess of Sussex officially opened the University of Chichester’s new Engineering and Digital Technology Park on 3rd October 2018. Part of the tour of the £35million development, based at the Bognor Regis campus, was a visit to the Department of Engineering and Design. The laboratory features several Mecmesin force measurement machines, notably a 50 kN OmniTest Universal Testing Machine (UTM); a state-of-the-art software-controlled materials testing system.
Mecmesin, a 41 year old British company specialising in the design and manufacture of force and torque testing equipment for quality control, has been acquired by PPT Holdings, a Battery Ventures company. Battery Ventures is a global investment firm focused on technology and innovation worldwide that has raised $6.9 billion since inception.
Mecmesin sponsors Millais School through the Learning Partnership
Mecmesin is sponsoring Millais School in Horsham, for the Race For The Line challenge run through The Learning Partnership. The initiative is intended to inspire tomorrow's engineers and scientists with a national competition involving around 70,000 students.
Mecmesin’s ABC-t Automated Bottle Closure Tester featured as the lead case study in the launch of Siemens Solid Edge 2019 3D CAD software at the PLM Connections event in Phoenix, Arizona. Tools available with SE out of the box are able to perform the Finite Element Analysis and volumetric calculations necessary to avoid having to manufacture expensive prototype parts.
Testing food packaging - quality assurance questions and answers
The packaging of food presents a unique challenge in the sector due to the conflicting objective requirements of functionality and legislation and the subjective, emotional and sensory judgement of aesthetics, ease of use and ultimately the product contained.
Testing beauty product packaging – ensuring functionality and form
Beauty may be more than skin deep, however brand reputation and loyalty in this market can very much depend on the outer packaging - failure to open, inform, dispense or re-seal with the expected qualities, may discourage the customer from experiencing the product within. For such a design-led industry, guaranteeing optimum functionality and form requires quantifiable test methods to measure these attributes.
Methods of force and torque testing for food packaging quality
Packaging solutions for food are incredibly diverse; in all aspects of physical design, mechanical function and aesthetic appeal. Their purpose is more straightforward: be safe, secure, yet easy to open, close or reliably re-seal; dispense cleanly and efficiently; minimise loss and waste; maximise shelf life; be cost-effective to manufacture and ultimately recycle or dispose of. Add to that, the consumer desire for attractive and innovative new designs, and the straightforward ultimate purpose can become quite involved in terms of meeting each individual challenge.
Mecmesin celebrates 40 years of testing packaging to perfection
The importance of visually-appealing packaging is obvious, but just as important are the quality, functionality and integrity of the utilised materials for consumer confidence and brand reputation. We caught up with Declan Tierney, Sales Director at Mecmesin, to learn about the key role their equipment plays in ensuring your packaging stands up to the test.
Motor-driven torque testing systems
Mecmesin can offer torque test stands to suit any requirement, from higher-torque PC-controlled systems, to touch-screen testers suited to production areas, high-precision torque testers, and more basic motorised stands for flexible use with a digital gauge and torque sensor.
Universal testing machine range (0.5 - 5 kN)
Mecmesin offers a wide range of universal testing machines (UTM) for tension and compression testing. At the top end, our MultiTest-i systems are driven by the powerful Emperor™ control and analysis software on a separate PC, while our production-level testers in the MultiTest-xt series offer touch-screen ease of use and rapid processing. All of these systems give fully programmed control and accurate, repeatable testing, with test libraries for instant recall.
Universal testing machine range (10 - 50 kN)
Mecmesin offers a wide range of universal testing machines (UTM) for tension and compression testing, including high loading for larger containers and packaging. At the top end, our MultiTest-i systems are driven by the powerful Emperor™ control and analysis software on a separate PC, while our production-level testers in the MultiTest-xt series offer touch-screen ease of use and rapid processing.
Mecmesin announces ‘Helixa’ precision torque tester
New from Mecmesin, the Helixa precision torque tester has been designed for achieving precise alignment when using the most sensitive torque sensors. It is a significant development in the torque testing of medical devices. Designers and engineers of the smallest rotary controls, the lightest springs, the finest bearings, and the most precise rotating connectors are already adopting the Helixa for its leading-edge features.
Mecmesin announces new friction, peel and tear tester
Leading testing instruments manufacturer Mecmesin has introduced its latest model, the FPT-H1 horizontal friction peel and tear tester. Designed for ease of use and supplied pre-programmed for all the main industry standard friction-testing methods for plastic films, paper and board. With fixtures also for peel and tear testing, it is ideally suited to the packaging industry, and for manufacturers, converters and users of sheet materials.
- Mike Webber
- Product Info
- Created 10 Apr 2017
- Modified 02 Aug 2017
- Hits 1279